切片机在开机前，要对限位开关、压力、安 全指示灯、气动安 全控制杆以及刀片挡板等进行检查，检查没有问题后才能接通电源，安 全联锁装置，要经常进行检查，是否能够正常工作。切片机运转时，不能进行清洁、检修及维护保养工作不能打开安 全防护门，移动光电管。如要进行这些工作，应切断设备电源，让设备处于断电状态，并挂上警示标志。
Before starting the slicer, check the limit switch, pressure, safety indicator, pneumatic safety control lever and blade baffle. The power can be connected only after there is no problem. The safety interlock should be checked frequently to see whether it can work normally. When the slicer is running, it is not allowed to carry out cleaning, overhaul and maintenance. It is not allowed to open the safety protection door and move the photocell. In order to carry out these works, cut off the power supply of the equipment, leave the equipment in the power-off state, and hang warning signs.
If the unloading position of the tipping arm of the slicer is jammed, the air pipe on the tipping arm shall be pulled out first, and then the inspection shall be carried out to eliminate the fault. After the equipment has a problem, it cannot be handled without authorization, and personnel shall be invited to handle it. When replacing the blade of the slicer, the operation shall be carried out according to the regulations. The pressure shall be confirmed to have been vented. After the operation is completed, the operation site shall be cleaned, the machinery and equipment shall be cleaned, and daily maintenance shall be carried out. Turn on the power supply of the slicer host, turn on the switch, and use the box temperature setting button to set the temperature required for slicing. At this time, the temperature display window will display the set temperature synchronously.
After setting, the actual temperature will be displayed after 5 seconds. At the same time, we can also check the set temperature at any time* During the first startup, a reference time shall be set through the reference time setting button, and the defrosting time shall be set through the defrosting time setting button. If manual defrosting is required, press the manual defrosting button first. After hearing the peak sound, press the quick refrigeration button to defrost the freezing table. Press the box temperature setting button to defrost the box. To close, just press again in this order.
The slicer can use the thickness adjustment knob to adjust the slicing thickness, and the slicing angle should be adjusted when installing the blade. Before slicing, cool the sample thoroughly, and then put it on the slicer. The sample should be close to the knife edge and adjust the plane. Then press the button to slice. The cut sample shall smoothly enter the gap between the anti rolling plate and the blade, Zui and then take it out for dyeing observation.
If you want to prevent the parameters of the slicer from being changed, you can lock the keyboard through the keyboard lock button on the control panel. Press the button for about 5 seconds. When the two points in the time window stop flashing, it means that the keyboard has been locked. The method to unlock is the same as above. Before starting the slicer again, check whether there is ponding in the slicer. If so, clean it and start it again. If the slicer needs to be used for a long time, its temperature at night can be adjusted below minus 10 degrees, which is conducive to the protection of the slicer.